VistaPure’s patented AutoFill accessories replenish SciCan STATIM® and BRAVO™ autoclaves with distilled-quality water automatically — enabling truly hands-free convenience.
VistaPure AutoFill accessory kits work with all VistaPure systems and can be installed easily on any STATIM 2000/G4 or STATIM 5000/G4, or any BRAVO autoclave, in just a few minutes!
- No more purchasing and storing distilled water
- No more cycle-fault errors caused by forgetting to fill reservoirs
- No more spills while filling autoclaves manually
- No more staff time spent filling autoclaves = more time with patients
VistaPure AutoFill Installation
A 12-foot length of clear polyurethane tubing is already attached to the AutoFill manifold. If the desired distance between the connection to the VistaPure and the autoclave is greater than 12 feet, simply order a longer length of the clear polyurethane tubing, as the distance from the VistaPure to the autoclave is not limited, except by the feed tubing.
Installing AutoFill on a STATIM
The VistaPure AutoFill for STATIM device (S3020) can be installed on any STATIM 2000/G4 or STATIM 5000/G4 autoclave as follows:
Installing AutoFill on a BRAVO
The VistaPure AutoFill for BRAVO device (S3030) can be installed on any model BRAVO autoclave.
In order to connect the VistaPure AutoFill to the BRAVO, one of the two tanks that come with the BRAVO must be selected as a “relay tank” since the BRAVO system uses a suction-lift pump to fill its reservoir when in its automatic-filling mode. Note that the relay tank must be located within 6 feet of the BRAVO unit, due to the suction-lift capacity of the BRAVO pump.
If using the small BRAVO tank:
If using the large BRAVO tank:
NOTE: If using the large BRAVO relay tank, you must install the included extension rod between the manifold and the float assembly.
Vista Research Group
PO Box 321
Ashland, Ohio 44805-0321